This True-RMS meter is feature rich, impeccably accurate, and competitively priced to meet the needs of any technician or tradesman. Measures AC/DC voltage, AC/DC current, Resistance, Capacitance and Frequency
I searched quite a while for a DMM that would measure a 30 MHz clock at 1.3V signaling. This one does it - even though this signal is well below the voltage level specified. The reported frequency is identical to what I measured on an Agilent o-scope. It is nice to have something so capable in a small footprint.To help ascertain its capabilities, it is unable to measure a 40Mhz signal at just under 1 V peak to peak - which is way below the voltage specification in the user guide. It had no problem with a 50 MHz 3.3V clock.It is rated to over 200 MHz at 5V signaling. I haven't tried to measure a clock that high or at that voltage level, but I have no reason to doubt it can do it.This is the ONLY meter I could find that would even spec frequency measurements this high.Build quality is awesome, the included software works, but is so-so.Today I was looking at low (1-50 uA) DC current measuring devices and I'm pleasantly surprised to see this meter has a DC current resolution down to 0.01 uA. I look forward to trying it out that functionality.Excellent value for the money. You can spend a lot more and get a lot less...
Incredible for the price. It comes with optical USB data cable, nice flexible soft test leads, a well thought out surface mount component adapter for smt resistors, diodes and devices (as in SOT-23, etc), and a 9 volt battery. When I removed the single screw at the top of the battery compartment, it felt like quality. Be careful when installing the 9 battery with the polarity. The battery mounts in the removable back piece which is then installed into the meter back and secured with a single screw.
However, my main message is that the catalog picture is misleading. The catalog image shows the '415 as a "0000" four digit display when "00000" (5 digits as required for a "22000" reading) is what you get.
Initial comparison with my trusty (and aged) Agilent 34401A (6 1/2 digits) are very similar. My guess is that there are some thermal issues (Seeback effect) with my bi-metallic connections. For under $100 , 22000 count resolution, and true RMS, this is a "steal."
Login Here to rate and review this product.
72-10405 - Professional Digital Multimeter with 4000 Count Display
This meter is feature rich, impeccably accurate, and competitively priced to meet the needs of any technician or tradesman. Measures AC/DC voltage, AC/DC current, Resistance, Capacitance, Frequency and temperature.
72-10410 - Professional TRMS Digital Multimeter with 6000 Count Display
72-10472 - K-Type Liquid Thermometer Probe
K-type probe is used for taking liquid temperature measurements. Needle immersion type.
50-14795 - Universal TV Stand, For Televisions 32" ~ 60"
Universal Television Stand
This universal TV stand is perfect for flat panel televisions that were mounted to the wall, but are to be moved to tabletop use. Universal mounting plate attaches to all VESA standard sets from 200x200 through 600x400, and provides a ...
35-5155 - 10MM Headphones White High Performance Earbud
High Performance Earbud Headphones White
Lightweight, stylish, high performance earbuds give crystal clear sound as well as a comfortable fit. Ideal for use with CD, MP3, MP4 and any other small portable audio players.
MCM Cat 57, Page 1220
MCM Cat 56, Page 1061
Requires Adobe Acrobat Reader.
MCM has always delivered an excellent service, ordered parts arrive on time and in excellent condition.